Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||19 August 2009|
|PDF File Size:||11.32 Mb|
|ePub File Size:||6.2 Mb|
|Price:||Free* [*Free Regsitration Required]|
The front feet contain foldaway tilt stands for convenience in bench operation. System components can be interconnected in virtually any configuration. Connect the A signal output to the Amplitude Each value should be within the calculated limits. Verify that ramp period variations do not exceed ten divisions.
HP A CALIBRATION MANUAL Pdf Download.
Set frequency to 99,9 Hz and function to square. The total quantity of a part is given the first time the part number appears. Frequency automatically changes to 10kHz.
Personnel should become familiar with the entire bulletin before beginning the calibration.
HP A User Manual | 23 pages | Also for: B
For these instruments, the following Amplitude Calibration adjustment procedure should 3325w used. The newer connectors are incompatible with the older cables as arc the newer cables incompatible with the older connectors.
Change oscilloscope horizontal control to.
Instruments in this range do not have A3R or A3C The information in this section documents the earlier instrument configurations and associated servicing procedures. Test instrument parameters Mannual distortion relative to fundamental at full output Amplitude modulation Square wave Ramp retrace Frequency Sine wave Square wave Triangle Positive slope ramp Negative slope ramp Phase offset Amplitude Amplitude accuracy with no manua, attenuator range 1 into manuap load no dc offset Flatness with no attenuation Auxiliary Dutput Sine Function Only.
Line 1 Because only numerical data can be placed in the variables, print statements may include in quotes the parameter interrogated. Without changing the reference level, set the A frequency and the spectrum analyzer center frequency to the frequencies listed below.
Each entry parameter car. It is necessary merely to enter the numerical data followed by the V or mV delimiter. 33325a instruments contain the amplitude control circuitry shown in Figure Change A function to triangle frequency will go to 10 kHz. If the program string were corrected to make all data valid, this printout would result from the above program.
Connect the A signal output to the second channel of the oscilloscope. Search the history of over billion web pages on the Internet. If the program string contains 48 characters or more, the A will hold up the bus while it processes the 48 characters before accepting and storing the rest of the string. This test verifies the sync output signal levels. Determine the value of a 0 using the equation: The horizontal arrow maual select the digit to be modified indicated by a bright digitand the vertical manal keys increment or decrement that digit.
Without changing the reference level, change the spectrum analyzer center frequency to To begin a single sweep: The following backdating information is organized by service group with applicable information placed together for easy reference. The power jack and mating plug of the power cable meet International Electrotechnical Commission IEC safety standards.
Providing the finest test equipment solutions since 1992
Replace the screw in the oven assembly and repeat Step e. Frame or chassis terminal. Note — the -hp- part number for U5 is The numbered delta in the text or on a schematic corresponds to the numbered delta shown in 3352a heading that precedes the backdating information for that particular service group. Connect A signal output to the voltage divider of Figure A.
Amplitude Control Circuitry Serial Rise and Fall Time: Adjust Offset In A3R33 for a voltmeter reading manuak 0. Calculate each value for y using the equation: Low-noise or noiseless, clean ground earth terminal. Maintenance by the operator is limited to cleaning or replacing the rear panel fan filter, or replacing the ac line fuse on the rear panel.
Manuap Supplies Service Group O NOTE A lighted indicator in the center of any entry key denotes it as the active entry parameter.